Nuclear Magnetic Resonance
From Physics 111-Lab Wiki
Nuclear Magnetic Resonance CW and Pulsed Description
- Note that there is NO eating or drinking in the 111-Lab anywhere, except in room 286 LeConte on the bench with the BLUE tape around it. Thank You the Staff.
In 1945 Felix Bloch (Stanford) and Edward Purcell (Harvard) discovered nuclear magnetic resonance in ordinary matter, for which they were awarded the Nobel Prize in 1952. This phenomenon has found many applications in science and technology, including magnetic resonance imaging used in medical practice.
In the NMR experiment, nuclear dipoles (the samples) are placed in a static magnetic field of about 4000 gauss, and in a time-varying radio-frequency magnetic field perpendicular to the static field. The static field causes Zeeman-effect splitting between sub-states, and the radio frequency field is tuned to the Larmor frequency so that it induces transitions between the sub-states. The resonance condition is observed using the Bloch two-coil induction technique. You will observe proton and fluorine nuclei. You will learn techniques of lock-in detection and signal averaging.
A second part of this experiment uses a pulsed radiofrequency field rather than a continuous-wave (CW) field. Signals are detected immediately after the pulsed excitation stops. The observable effects are comparable to the free vibration or ringing of a resonant cavity on an atomic scale. This is the basis of Magnetic Resonance Imaging in the medical field today
- Pre-requisites: Physics 137B
- Days Alloted for the Experiment: 9
- Consecutive days: No
I. Pulsed NMR
Reprints and other information can be found on the Physics 111 Library Site
This lab will be graded 30% on theory, 40% on technique, and 30% on analysis. For more information, see the Advanced Lab Syllabus.
Comments: E-mail Don Orlando
Before the Lab
Complete the following before your experiment's scheduled start date
- View the two videos CW NMR and Pulsed NMR.
- Complete the NMR Pre Lab and Evaluation sheets. Print and fill it out. The Pre-Lab must be printed separately. Discuss the experiment and pre-lab questions with any faculty member or GSI and get it signed off by that faculty member or GSI. Turn in the signed pre-lab sheet with your lab report.
- View the Transitions Lecture, Transitions
- Bloch, Felix "Nuclear Magnetism"; American Scientists: Vol. 43, No. 1, Jan 1955, pp. 48-62.
- C. Kittel, Introduction to Solid State Physics, 5th ed., John Wiley (1976). Read pp. 479-509 for a brief-quantitative expose of the main ideas. #QC176.K51
- L Yuan and C. S. Wu, Methods of Experimental Physics, Part B, Vol. 5, Academic Press (1963), pp. 104-123 (Section 126.96.36.199). This reference discusses all the ideas necessary to do the experiment, which uses the two-coil Bloch method. #QC33.M48
- F. Bloch, "Nuclear Induction", Physical Review 70, 460. No 7 and 8, (1946). Bloch's two-coil method is used in this experiment.
- R. Schumaker and W. A. Benjamin, Introduction to Magnetic Resonance, 1970. Read Ch. 2 and Ch. 3. #QC762.S34
You should keep a laboratory notebook. The notebook should contain a detailed record of everything that was done and how/why it was done, as well as all of the data and analysis, also with plenty of how/why entries. This will aid you when you write your report.
In 1952, Felix Bloch and Edward Purcell received the Nobel Prize in physics for their discovery of nuclear magnetic resonance in 1945. Bloch's method of observation is now widely used in many areas of science and technology. NMR is a sensitive probe for determining the local magnetic field at the location of the nuclei in matter. It gives us information about nuclear spins and their surroundings. In the medical field, it is called Magnetic Resonance Imaging to avoid the use of the word "nuclear". This experiment has the following objectives:
- To observe the phenomena of nuclear magnetic resonance (NMR) of protons in H2O and other liquids
- To observe the absorption and dispersion line shapes of NMR under slow passage and under non-adiabatic passage conditions, and to study their dependence on the concentration of paramagnetic ions added to the liquid
- To measure the ratio of the magnetic moment of F19 to that of the proton
- To use the technique of lock-in detection to get improved signal-to-noise ratios in the NMR detection.
A former student commented, "The best thing that can be said about NMR is that, in beginning of the experiment, a student feels overwhelmed by its complexity. As you become more familiar with it, this feeling is replaced by curiosity and eventually understanding." This is to say that while there is a great deal to learn initially about NMR, given time and a lot of effort the pieces do fall into place, and the result is an unusually rewarding experiment.
Conceptually this experiment is quite simple, but how the data are recorded is not. It helps to know what we are going to do before we discuss what pieces of equipment we use.
Suppose we place a single nucleus between the south and north poles of the magnet. The interaction between the magnetic moment of the nucleus with the local magnetic field creates equal and opposite forces that form a torque upon the nucleus in the classical picture. The magnetic moment then begins to rotate about the vertical axis. This rotation is called Larmor "precession". The frequency of this precession is called the Larmor frequency. In this experiment we will place the sample of protons in a permanent magnet. This will force the protons to precess at the Larmor frequency and induce an electric field in a receiver coil that is wrapped around the sample. At the same time we will apply a radio frequency to the sample. This frequency will be tuned to the precession frequency, yielding a resonance. Our goal is to observe the nuclear magnetic resonance of the protons in the sample.
In the quantum mechanical picture, a sample of protons in a magnetic field of strength H0 has energy levels that are populated according to the Boltzmann distribution. When we send in photons (electromagnetic radiation) of energy the system absorbs some of the photons. This frequency, known as the Larmor frequency, is the resonance frequency of the system. At the resonance, the gyromagnetic ratio can be determined from the ratio . Alternatively, once we find the resonance frequency, we can determine H0 using the known value of .
A typical resonance curve is shown in Figure 1. How shall we produce this curve so that we can measure H0 at resonance? We could set to a fixed value and make measurements of (power out)/(power in) for many different values of H, in small enough increments, to determine H0. Of course we could also set H to a fixed value, and make measurements as is varied. We really don't care about the exact amount of power - we need only the ratios - but we do need some way of measuring the power, which in our case is at a frequency of about 16 MHz.
In the lab, we shall scan the applied magnetic field H back and forth relatively slowly above and below the resonance value Ho. We will use a scanning frequency of 60 Hz. The variation of H in time will look like Figure 2 when we observe it with an oscilloscope.
When Figures 1 and 2 are put together, we obtain Figure 3.
Next is to visualize the resonant "signal" shown in Figure 1. If the input power is kept constant, the resonant signal can be realized as a change in the output power. We will consider the electric field, rather than the power, of the outgoing radiation at the frequency fo as a function of time and call it the signal. The amplitude of this signal is a measure of the power absorbed in the sample.
The incident electric field at a frequency of 16 MHz shown in Figure 4a is now modulated in amplitude at a frequency of 60 Hz as illustrated in Figure 4b. The magnitude of the amplitude when plotted in time is a measure of the resonance curve. The function of the detector is to rectify the 16 MHz signal by putting it through a low-pass filter that removes the 16 MHz component but leaves the modulating component at 60 Hz. The resulting signal is sent to an oscilloscope with the sweep frequency set at 60 Hz. If we operate the oscilloscope in the x-y mode, with the x-channel being the modulating magnetic field and the y-channel being the filtered signal, the display of the oscilloscope is the desired resonance curve.
A block diagram with signals at the various stages will be shown in the next section.
To summarize, we place a test tube of a liquid sample in a magnetic field. We then apply electromagnetic radiation at radio frequency to the sample with a nearby transmission coil. We observe the resonance at the Larmor frequency by tuning the input frequency. As the protons precess, they induce an emf in a receiver coil wrapped around the sample. The amplitude of the induced field is at a maximum when the frequency of the applied RF field exactly matches the Larmor frequency. In the detecting circuit we rectify the RF field, filter it to get a DC voltage proportional to the amplitude of the field, and display it on an oscilloscope.
To take data, we set the frequency of the applied RF to the Larmor frequency for the static field produced by the permanent magnet. By superimposing a modulating magnetic field, we then sweep the amplitude of the resulting magnetic field sinusoidally in time at 60 Hz, above and below the static field for resonance (we say that the field is modulated sinusoidally with a frequency of 60 Hz). The final result is a plot of the resonance curve, the amplitude of the detector signal vs. the magnetic field applied to the sample, for a fixed frequency.
In the first part of this experiment we are going to use a Ho about 3.9 kG permanent magnet. The coils of wire wrapped around the poles of the permanent magnet are for the purpose of varying the field at a very low frequency and are needed in the later part of the experiment, where a lock-in amplifier is used.
It is also used with the Lock-in and function generator for sweeping the magnetic field. This is done by using the magnetizing coils wrapped around the magnet iron core.
Details of the NMR head are shown in Figure 7. It is a brass box containing a radio-frequency transmitting coil positioned perpendicularly to a receiving coil into which a test tube contain a sample of protons in H2O or other liquids is inserted. The NMR head is placed between the poles of the permanent magnet. Do not remove this head from the magnet, nor disconnect its cables. Instead, you may examine a spare NMR head on the table.
The top and bottom covers of the NMR head have 7-turn pancake coils which carry varying currents that produce a modulation field , where is 60 Hz. The amplitude is controlled from the modulation unit with gauss/amp coming from the modulating current. This is done with a 1.7 amp 60Hz power supply to modulate the field.
This black box NMR RF Black Box Circuit Diagram is permanently mounted on the magnet stand and is not to be removed or disassembled; do not remove the cables either. It contains a tunable RF oscillator centered around 16.54x,xxx Mhz, a tunable receiver (left-hand knob), a diode-based detector (right-hand knob), a low-pass filter, and an amplifier for enhancing the detector output. Refer to 'NMR RF Box Circuit Diagram' in CW Appendix I for details. Record this frequency down to 1 Hz resolution for the pulse NMR section.
The oscillator generates an RF signal that produces a magnetic field in the sample. The frequency is determined by where L and C are the combined inductance and capacitance of the coil in the NMR head, the variable capacitors in the NMR box, and the cables connecting the two. One of the capacitors tunes the circuit to hit resonance at the Larmor frequency. Because the resonant circuit includes inductors in the head and in the cables, as well as capacitors in the NMR box and the cables, touching or moving anything during measurements makes the frequency unstable and introduces lots of noise. Thus, be gentle, and keep your hands off while taking data.
The amplitude of the RF field H1 is controlled by the DC supply voltage V1 to the oscillator. The direction of H1 lies in a plane perpendicular to the DC field Ho of the large permanent magnet. However, the direction relative to the axis of the receiving coil can be adjusted by rotating a copper disk on a "paddle," which controls the phase of the leakage voltage into the receiving coil. Changing the phase enables us to observe either absorption, dispersion or a mixture of the two. In this lab, we will observe adiabatic absorption or dispersion, and non-adiabatic absorption or dispersion modes. See Figures 7A, 7B, and 7C for examples of these modes.
Refer to Figure A for the NMR Block Diagram:
The NMR head is placed in the DC field gauss of a large permanent magnet. The simplest method for observing NMR is to sweep the magnetic field through resonance at 60 Hz by superimposing a modulating Hmod on H0. Also, it is helpful to display the NMR signal Vac as the y-input of an oscilloscope and the 60-Hz phase shifted sine wave from the modulation unit as the x-input.
The signal from the NMR box is amplified by the PAR 113 amplifier which is used to limit the upper and lower pass frequency (typically 1 kHz and 3 Hz), thereby rejecting unwanted frequencies and improving the visibility of the desired signal. The Fluke 1953A frequency counter is used to read the NMR oscillator frequency without perturbing the oscillator.
The coils wrapped around the pole pieces of the magnet were initially used to magnetize the Alnico alloy of the poles, but are no longer needed or used for this purpose. Instead, they are used in the second part of the experiment for superimposing on Ho a small monotonically increasing field H2(t) generated by the Function Generator (a "ramp" field). The field reaches a maximum amplitude of about 60 gauss. The ramp frequency can be as small as 0.01 Hz, thereby enabling slow, repetitive scanning in one direction through the magnetic resonance. The Lock-in amplifier (see Appendix) is used for recording the derivative of the NMR signal.
To observe proton resonance, use the prepared set of samples of H2O + MnCl24H2O with yellow mylar tape on the top. The Mn molarity (M) ranges from M = 3.3, 1.0, 0.33, ..., to 10-5 moles of MnCl24H2O per liter of solution, and gives a wide range of relaxation times. A one molar sample contains 19.7 grams of MnCl24H2O in 100 cc of mixed solution. Other useful samples are glycerin with a small amount of FeCl3, and pure H2O. For F19 we are going to use Teflon rods. Samples should be tightly stopped and contain no air bubbles. Be careful not to drop or open the test tubes. Call for the staff if any solution comes out of the test tubes.
NMR Lock In Amplifier
Part of Lock-In Manual model SRS 830
Lock-in amplifiers are used to detect and measure very small AC signals - all the way down to a few nanovolts! Accurate measurements may be made even when the small signal is obscured by noise sources many thousands of times larger.
Lock-in amplifiers use a technique known as phase-sensitive detection to single out the component of the signal at a specific reference frequency AND phase. Noise signals at frequencies other than the reference frequency are rejected and do not affect the measurement.
Full Manual Lock-in
Why use a lock-in?
Let's consider an example. Suppose the signal is a 10 nV sine wave at 10 kHz. Clearly some amplifi-cation is required. A good low noise amplifier will have about 5 nV/ of input noise. If the amplifier bandwidth is 100 kHz and the gain is 1000, then we can expect our output to be 10 μV of signal (10 nV x 1000) and 1.6 mV of broadband noise (5nV/ x kHz x 1000). We won't have much luck measuring the output signal unless we single out the frequency of interest.
If we follow the amplifier with a band pass filter with a Q=100 (a VERY good filter) centered at 10 kHz, any signal in a 100 Hz bandwidth will be detected (10 kHz/Q). The noise in the band pass filter will be 50 μV (5 nV/ x Hz x 1000 ) and the signal will still be 10 μV. The output noise is much greater than the signal and an accurate measurement can not be made. Further gain will not help the signal to noise problem.
Now try following the amplifier with a phase sensitive detector (PSD). The PSD can detect the signal at 10 kHz with a bandwidth as narrow as 0.01 Hz! In this case, the noise in the detection bandwidth will be only 0.5 μV (5 nV/ x Hz x 1000) while the signal is still 10 (V. The signal to noise ratio is now 20 and an accurate measurement of the signal is possible.
What is phase-sensitive detection?
Lock-in measurements require a frequency reference. Typically an experiment is excited at a fixed frequency (from an oscillator or function generator) and the lock-in detects the response from the experiment at the reference frequency. In the diagram below, the reference signal is a square wave at frequency (r. This might be the sync output from a function generator. If the sine output from the function generator is used to excite the experiment, the response might be the signal waveform shown below. The signal is
where is the signal amplitude.
The SR830 generates its own sine wave, shown as the lock-in reference to the right. The lock-in reference is .
The SR830 amplifies the signal and then multiplies it by the lock-in reference using a phase-sensitive detector or multiplier. The output of the PSD is simply the product of two sine waves.
The PSD output is two AC signals, one at the difference frequency and the other at the sum frequency .
If the PSD output is passed through a low pass filter, the AC signals are removed. What will be left? In the general case, nothing. However, if ωr = ωL, the difference frequency component will be a DC signal. In this case, the filtered PSD output will be
This is a very nice signal - it is a DC signal proportional to the signal amplitude.
Narrow band detection
Now suppose the input is made up of signal plus noise. The PSD and low pass filter only detect signals whose frequencies are very close to the lock-in reference frequency. Noise signals at frequencies far from the reference are attenuated at the PSD output by the low pass filter (neither are close to DC). Noise at frequencies very close to the reference frequency will result in very low frequency AC outputs from the PSD (). Their attenuation depends upon the low pass filter bandwidth and roll-off. A narrower bandwidth will remove noise sources very close to the reference frequency, a wider bandwidth allows these signals to pass. The low pass filter bandwidth determines the bandwidth of detection. Only the signal at the reference frequency will result in a true DC output and be unaffected by the low pass filter. This is the signal we want to measure.
Where does the lock-in reference come from?
We need to make the lock-in reference the same as the signal frequency, i.e. ωr = ωL. Not only do the frequencies have to be the same, the phase between the signals can not change with time, otherwise will change and Vpsd will not be a DC signal. In other words, the lock-in reference needs to be phase-locked to the signal reference.
Lock-in amplifiers use a phase-locked-loop (PLL) to generate the reference signal. An external reference signal (in this case, the reference square wave) is provided to the lock-in. The PLL in the lock-in locks the internal reference oscillator to this external reference, resulting in a reference sine wave at ωr with a fixed phase shift of θref Since the PLL actively tracks the external reference, changes in the external reference frequency do not affect the measurement.
All lock-in measurements require a reference signal.
In this case, the reference is provided by the excitation source (the function generator). This is called an external reference source. In many situations, the SR830's internal oscillator may be used instead. The internal oscillator is just like a function generator (with variable sine output and a TTL sync) which is always phase-locked to the reference oscillator.
Magnitude and phase
Remember that the PSD output is proportional to Vsigcosθ where . θ is the phase difference between the signal and the lock-in reference oscillator. By adjusting θref we can make θ equal to zero, in which case we can measure Vsig . Conversely, if θ is 90°, there will be no output at all. A lock-in with a single PSD is called a single-phase lock-in and its output is Vsigcosθ.
This phase dependency can be eliminated by adding a second PSD. If the second PSD multiplies the signal with the reference oscillator shifted by 90°, i.e. , its low pass filtered output will be
Now we have two outputs, one proportional to cosθ and the other proportional to sinθ. If we call the first output X and the second Y,
X = Vsigcosθ Y = Vsigsinθ
these two quantities represent the signal as a vector relative to the lock-in reference oscillator. X is called the 'in-phase' component and Y the 'quadrature' component. This is because when θ=0, X measures the signal while Y is zero.
By computing the magnitude (R) of the signal vector, the phase dependency is removed.
R measures the signal amplitude and does not depend upon the phase between the signal and
A dual-phase lock-in, such as the SR830, has two PSD's, with reference oscillators 90° apart, and can measure X, Y and R directly. In addition, the phase ( between the signal and lock-in reference, can be measured according to
WHAT DOES A LOCK-IN MEASURE?
So what exactly does the SR830 measure? Fourier's theorem basically states that any input signal can be represented as the sum of many, many sine waves of differing amplitudes, frequencies and phases. This is generally considered as representing the signal in the "frequency domain". Normal oscilloscopes display the signal in the "time domain". Except in the case of clean sine waves, the time domain representation does not convey very much information about the various frequencies which make up the signal.
What does the SR830 measure?
The SR830 multiplies the signal by a pure sine wave at the reference frequency. All components of the input signal are multiplied by the reference simultaneously. Mathematically speaking, sine waves of differing frequencies are orthogonal, i.e. the average of the product of two sine waves is zero unless the frequencies are EXACTLY the same. In the SR830, the product of this multiplication yields a DC output signal proportional to the component of the signal whose frequency is exactly locked to the reference frequency. The low pass filter which follows the multiplier provides the averaging which removes the products of the reference with components at all other frequencies.
The SR830, because it multiplies the signal with a pure sine wave, measures the single Fourier (sine) component of the signal at the reference frequency. Let's take a look at an example. Suppose the input signal is a simple square wave at frequency 'f'. The square wave is actually composed of many sine waves at multiples of 'f' with carefully related amplitudes and phases. A 2V pk-pk square wave can be expressed as
where . The SR830, locked to 'f' will single out the first component. The measured signal will be , not the 2V pk-pk that you'd measure on a scope.
In the general case, the input consists of signal plus noise. Noise is represented as varying signals at all frequencies. The ideal lock-in only responds to noise at the reference frequency. Noise at other frequencies is removed by the low pass filter following the multiplier. This "bandwidth narrowing. is the primary advantage that a lock-in amplifier provides. Only inputs at frequencies at the reference frequency result in an output.
RMS or Peak?
Lock-in amplifiers as a general rule display the input signal in Volts RMS. When the SR830 displays a magnitude of 1V (rms), the component of the input signal at the reference frequency is a sine wave with an amplitude of 1 Vrms or 2.8 V pk-pk.
Thus, in the previous example with a 2 V pk-pk square wave input, the SR830 would detect the first sine component, . The measured and displayed magnitude would be 0.90 V (rms) .
Degrees or Radians?
In this discussion, frequencies have been referred to as f (Hz) and ( radians/sec). This is because people measure frequencies in cycles per second and math works best in radians. For purposes of measurement, frequencies as measured in a lock-in amplifier are in Hz. The equations used to explain the actual calculations are sometimes written using ( to simplify the expressions.
Phase is always reported in degrees. Once again, this is more by custom than by choice. Equations written as are written as if is in radians mostly for simplicity. Lock-in amplifiers always manipulate and measure phase in degrees.
The Functional SR830
The functional block diagram of the SR830 DSP Lock-In Amplifier is shown below. The functions in the gray area are handled by the digital signal processor (DSP). The DSP aspects of the SR830 are covered in the SR830 manual. They are described as they come up in each functional block description. ( See Section three (3) in the SR 830 Manual)
This experiment uses a Digital Storage oscilloscope in order to examine the weak resonance signals. See the included manual located in the reprints on how to use the digital storage scope. The output signal is also connected to the computer so that the data could be transferred to the computer and stored for later use. The instructions on how to transfer the data to the computer are located in the CW Appendix IV.
- HEATHKIT POWER SUPPLY. Flip the POWER switch to STANDBY and wait 1 minute; then switch to ON. Set the METER SWITCH to the RIGHT. Set the amplitude of the RF field by adjusting the B+ OUTPUT knob to 150 volts, as read by the voltmeter.
- On the Hmod magnetic field modulation control panel. Flip POWER and SWEEP ON switches up. Turn the AMPLITUDE ADJUST knob to its maximum CW position to maximize the modulation of the field. This sends about 1.7 amp at 60 HZ to the modulation coils; this current generates a magnetic field of about 2.9 gauss.
- The PHASE ADJUST knob on the Hmod panel changes the phase of the modulation signal sent to the x-input of the scope relative to the signal sent to the modulation coils, and hence relative to the detected output signal sent to the y-input of the scope. In short, it changes the phase between the x and y signals. No need to set it at this time.
- PRE-AMP, Stanford Research Systems Model SR 560. Turn the unit on and plug the AUDIO OUTPUT from the back of the NMR box into INPUT A and make sure that the DC /GND/AC switch next to the A input is in the AC position. Note that if you suddenly lose your original signal while you are making various adjustments - if your scope trace goes flat - you should push the OV LD overload recovery switch down. Set the gain to 50, the LF ROLL-OFF to 3 Hz, and the HF ROLL-OFF to 10K. You will probably have to adjust these later to make your signal as clean as possible without losing any of its major features.
- Turn on the RF POWER SUPPLY.
- Turn on the frequency counter (Fluke PM6669) and the oscilloscope. Connect the FREQUENCY MONITOR on the back of the NMR box to the input of the frequency counter. Put the scope in the x-y mode. Then connect the PHASE ADJUST to the x-input of the scope and the PRE-AMP output to the y-input of the scope.
- Now you're ready to go. Insert a glycerin sample into the NMR HEAD and find the resonance signal by doing the following. Look up in the NMR Frequency TABLE of the manual - the NMR frequency for H1 in a 10 kilogauss field. Knowing this number and that our Ho is about 3.9 kG, calculate the approximate NMR frequency for our set-up (around 16.54xxx). Turn the TRANS FREQ knob on the NMR BOX until the counter reads this value. Now adjust the receiver coil to observe the resonance by adjusting the CR control so that the receiver frequency matches that of the transmitter. Start with it at the fiducial mark.
Slowly vary the frequency around the value you calculated above. You should be able to find the resonance, and your signal should look like Fig. 8. Note that the resonance condition is very sensitive to frequency; so go slowly. Once you have found it, "peak-up" your signal by readjusting CR on the NMR BOX for maximizing the signal. If you don't find a signal after you've fussed with everything, ask for help.
Move the NMR HEAD around gently in the gap until you have maximized the number of "wiggles" and minimized the line width. This will place the head in the most uniform or homogeneous region of the magnet, and you want to keep it there. Once you have set the position for the day, don't change it. Some of your calculations depend on the field being the same for successive measurements. Repeat this procedure each day when you first come to the lab, since other people may have moved the NMR head from where you have determined to be the best.
8. Read over the REPORT section to be sure you take all the necessary data. Read the information described in the CW Appendix IV on how to transfer data to the computer.
A. By using the glycerin sample you can find a narrow line (that is, a sharp resonance) in the homogeneous region of the magnet. Once the resonance is observed, tune the frequency of the RF, f, until the two peaks are symmetric about the midpoint of the Hmod trace on the scope, for both slow passage and non-adiabatic passage conditions. Measure the resonant frequency and hence (by knowing γ) the magnetic field to high precision. It is a standard technique to use NMR to determine the magnetic field strength precisely.
B. By recalling that , where f is the frequency and H is the magnetic field, you can devise a method of calibrating (in gauss/div) the field axis of the oscilloscope. Then you can transfer the data of the four modes (absorption/dispersion; slow passage/non-adiabatic passage) to the computer.
C. Replace the glycerin sample with a 0.01M Mn++ sample; again this signal should look like Figure 8. Download the data of this non-adiabatic rapid-passage absorption mode to the computer. (See REPORT, #3.) Now insert a more highly doped sample, 1M Mn++. The paramagnetic Mn++ ions relax the proton spins and the signal is closely resembled the one of a slow passage absorption mode. By further rotating the paddles you should be able to get a signal like Fig. 10, approximately a slow passage dispersion mode. You may have to increase V1 to see this.
D. For observing the F19 resonance we use a Teflon rod. Note that the F19 resonance is a little harder to find. Using the same procedures as for H1, find the resonance curves for F19. Accurately measure the NMR frequency ratio of F19 to H1. To do that, measure f(H1) and f(F19) each a number of times under identical conditions.
E. Lock-in Amplifier Operation: For fine tuning the NMR resonance, we can leave the frequency f and all the other parameters fixed and optimized, and vary the "total-DC" magnetic field by adding a small field H2 with the output of the Wavetek 164 generator driving the wire coils. The following diagrams show what this change does to the signal. See Figure 11.
Read "Basic Introduction to Lock-in Amplifiers" in the Lock-in Amplifier, and the manual of the Stanford Research Systems Model SR 830DSP Lock-in Amplifier (copied in reprints). In this apparatus, the lock-in amplifier records the derivative of the NMR signal as the frequency sweeps slowly through the resonance by ramping the field H2 with the Wavetek generator. The signal-to-noise ratio can be 100x larger than that observed with the oscilloscope. This method is used to find and record weak signals. It also provides a convenient way to record the line width δH of a signal.
As usual, getting started is not trivial. First you need to have a good signal on the oscilloscope. Then turn on the power supply of the magnet, the lock-in amplifier, and the Wavetek generator. Set the Wavetek generator to give a .03 Hz triangular wave with maximum amplitude. If you are not familiar with the Wavetek generator, borrow another oscilloscope and look at the output signal, and the effects of turning or switching the knobs. Check out the OFFSET control.
You want to sweep the field as shown in Figure 11. Unfortunately, the power supply of the magnet does not generate a negative current and the sweep starts at the resonance field Ho. Consequently, the sweep is not symmetrical, and half the resonance curve is lost. Therefore you must change the RF frequency to a higher value to put the resonance at the middle of the triangular sweep. This means you must put the resonance curve on the scope at one end rather than in the center at a larger frequency than used in the previous part of the experiment. How much? Experiment to find out.
Connect the output of the pre-amplifier to the A-input of the lock-in amplifier. Set the lock-in controls as follows: Phase - 0; zero offset, switch off; Time Constant - 1 sec. Connect phase input (ref) to the phase adjust on the Hmod panel. Connect output of the lock-in to the Y-input of the PULSE NMR DAQ INTERFACE BOX and switch the box to CW.
Now how do you start? Turn the amplitude of Hmod way down, almost to zero. Refer to the diagrams above. For the first part of the experiment, Hmod was large; for this part it must be small. But of course once you get things working, try twisting all the knobs to peak up the signal. You should see the resonance curve sweeping across the face of the scope, and off the end. A few seconds later it should return and sweep across and off the other end. The voltage on the magnet power supply should be going from zero up and back again. If you are monitoring the Wavetek output on another oscilloscope, the horizontal line should go up and down from zero. When it is halfway, the resonance curve should be in the center of the scope.
For the H2O samples with molarity of 3, 1, 0.3, 0.1, 0.03 molar Mn++, adjust the mode to absorption and set Hmod small, ~1/8 the line width; this takes the derivative (see Ref. 2, p. 51). Adjust the lock-in amplifier and record the derivative with the computer.
Figure 12 shows a typical result for a 1 Molar sample. The two sharp glitches occur at the ramp voltage rapid reset. (Chart speed = 5 cm/minute; Wavetek: q15V at 0.01 Hz; lock-in time contact = 0.3 sec; sensitivity = 25 mV on channel A input; DC pre-filter = 100 ms; input filters: hi-pass = 50 Hz, lo-pass = 0.1 kHz; mode = fund f; lock-in reference phase adjusted to give maximum peak-to-peak signal on lock-in meter; output zero offset adjusted to center trace on paper.) By scanning more slowly, one can get a good measure of the line width H between the peaks. To measure the signal-to-noise ratio (S/N), one can increase the input gain by, say, 10x or more to record the RMS noise off resonance.
When the NMR paddle is set to the dispersion mode for a 0.33 Molar sample, the lock-in signal is shown in Fig. 13 under very similar conditions to Fig. 12.
NMR Scope Program
How to transfer data from the scope to the computer.
First we need to make a connection between the computer and the oscilloscope. To do that, locate the aluminum box labeled Computer Interface Unit. Connect the X input (X/Trig CW) with the knob on the Hmod corresponding to the sign "to oscilloscope". Connect the Y input (Y In) with the output of PRE-AMP. The other port of the interface unit should have a 50-ohm termination. Don't forget to turn on the Computer Interface Unit.
Now we are ready to use the computer to transfer the data.
NMR Scope Program
The program we are going to use to transfer the data is called NMR Scope. To get familiar with the program you might want to follow the following simple exercise.
On the desktop you will find an icon NMR Scope. Double click on that icon to open up the program. You will see the following window:
- First select the type of experiment you are doing. There are two options: Pulsed NMR Rising Edge Triggering and CW NMR Falling Edge Triggering
- At the bottom left corner you will see three buttons: Display TY, Display XY, and Display Strip Chart. The latter one will be used later in the experiment. For now, press either Display TY or Display XY button.
- To see the signal press the TY Mode or the XY Mode, depending on the previous choice you have made. You should now see a signal in the upper right chart. This is what it might look like:
4. To stop the signal, press Quit TY, or Quit XY signal accordingly.
5. To save the data, press Save Current TY/XY Traces. Save the data as a .dat file format.
6. To clear the graph, press Clear TY/XY Traces.
Note: you can actually copy the graph and print it separately. Right click anywhere on the graph and press Copy Data. This temporary stores the image on the computer and you can paste it in most programs such as word or excel.
Averaging the Data:
We can also "average out" the signal to get a smother line. The program takes multiple samples over time and takes the average. You can average the data in wither mode: TY or XY. After choosing the mode, you might want to go over the following steps:
- Press TY/XY Average. The window comes up asking if you want to "AutoSave the data." If you wish to do so, press yes and it will prompt you to save it. Otherwise, press no. In either case, you will get a window, asking to "Chose Acquisition Points." Just press ok. In the middle to the left of the program window you can set either the acquisition time or acquisition points. By default they are set to maximum.
- Now, press Start TY/XY Average. The top right graph is the same as one before. The bottom right graph is the average graph. You might see similar graphs to the ones bellow:
3. To stop the signal, press Quit TY/XY Average.
4. To save the data press Save Averaged TY/XY Data.
5. To clear the graphs press Clear TY/XY Traces.
Strip Chart Recording Function (used in the later part of the NMR experiment)
Instead of using the actual Strip Chart Recorder, we can use the NMR Scope program to accomplish the same results. Once all the equipment has been set up properly, the use of the program is simple.
- Set up the equipment as instructed in the section "Taking Data," part E.
- Open the NMR Scope program.
- Press the button Display Strip Chart. You will see that you are only given one graph space in the top right corner.
- To start taking the data, press Start Chart. You will see something similar to the following graph:
- To stop the signal press Stop Chart
- To save the data press Save Strip Chart
- To clear the data press Clear Strip Chart
This concludes the overview of the program. If you have any more questions, you can ask one of the staff members for help.
Include the following data, analysis, and calculations.
- The Bloch Equations are the key to understanding this experiment:
- What is M? Is H a magnetic field inherent in your sample, or is it an applied field? Derive equations (1) to (3) for the case of no damping (none of the T1 or T2 terms). Hint: consider the classical torque equation N = dL/dt.
- The equations you derived are applicable to a set of identical magnetic moments (spins), i.e., all the spins see the same magnetic field. The damping terms in (1) to (3) are added to take the different environment of each spin into account.
- In equations (1) and (2), T2 is often called a dephasing time, the time required for (M)x,y to decay to zero after the resonance condition is removed. What is getting out of phase? How can this arise from inhomogeneities in H?
- T1 is the relaxation time for the z-component of M to come to an equilibrium value Mo when the resonance condition (applied RF) is removed. (You can see this by setting H to zero in (3).) What determines this equilibrium value, and what is it? [Hint: look at the classical Boltzmann distribution.]
- What (very general) physical factor of your sample accounts for the magnitude of T1? [Hint: Mo is a thermal equilibrium magnetization.] More specifically, how might the addition of paramagnetic ions Mn++ affect T1 (qualitatively)?
- Measure the magnetic field H0 of the magnet as precisely as you can in Gauss (average value and standard deviation). Are paramagnetic corrections significant? Measure H0 on several different days: does it vary? Why?
- Include best computer plot with calibrated H axes for proton signals in glycerin, showing the "best wiggles." Estimate the magnetic field inhomogeneity from the damping rate of the wiggles.
- Computer plots with calibrated H axes for absorption and for dispersion NMR signals in H2O under slow passage and under non-adiabatic passage.
- Measure as precisely as you can the ratio of the magnetic moment of F19 to H1 , with standard deviation. Show best scope photo of F19 resonance.
- Record and show sequence of absorption lock-in trace from the computer for protons in the presence of M = 3, 1, 0.3, 0.1, 0.03 Molar Mn++. Measure the line width (H in Gauss and plot (H vs. Molarity; explain this graph; what is the residual line width (in Gauss) due to field non-uniformity?
- From your best lock-in data for resonance in 1 Molar Mn++M in H2O, compute the expected S/N ratio for the deuteron resonance in a 1 Molar Mn++ in D2O sample under two different assumptions:
- The field Ho is kept at 3900 G and the oscillator frequency is adjusted to f(H2).
- The oscillator frequency is kept at 16.5 MHz and the DC field is adjusted for resonance.
- Under condition 2 above, compute the S/N ratio expected for O17 NMR resonance in naturally abundant water; compare to Ref. 5, Fig. 4.
Copyright (c) 2011 The Regents of the University California. All Rights Reserved
- A. Abragam, Principles of Nuclear Magnetism, Oxford Press, 1961. This is the definitive reference. #QC762.A23
- R. Liboff, Introduction Quantum Mechanics, Holden Day, 1980. Sections 11.8 and 11.9 include a simple, direct derivation of NMR, along with a physical interpretation. While Liboff does not use the classical Bloch equations, he does give a good idea of what's going on. #QC174.12.L54
- S. Walker and H. Straw, Spectroscopy Vol. 1, Macmillian (1962, Ch. 5. (Available in 111 Lab, not to be taken out of the laboratory). #QC451.W2 (Engineering Library)
- Bloch, F., et al "The Nuclear Induction Experiment"; Physical Review: Vol. 70, Oct 1946, pp. 474-485.(Note: Note check out CRC Table of Physical Constants for valves) Searchable Page
- Ames, D.P. "Magnetic Resonance": Chapter 9, pp. 8.112-8.127.
- Hahn, E.L. "Free Nuclear Induction"; Physics Today, Nov. 1953, pp.4-9.
- Brewer, Richard G. & Hahn, Erwin L. "Atomic Memory"; Scientific American: Vol. 251, No. 6, Dec. 1984, pp. 50-57.
- Lowe, I.J., & Whitson, D. "Simple Pulsed Nuclear-Magnetic Resonance Spectrometer"; pp. 335-338.
- Cohen-Tannoudji, "Quantum Mechanics"; Vol. 1, 1977, Wiley. pp. 443-454.
- Jacobsohn, Boris A. and Wangsness, Roald "Shapes of Nuclear Induction Signals"; Physical Review: Vol. 73, May, 1948, pp. 942-946. Searchable Page
- Bloembergen, N, Purcell, E. M., Pound, R.V. "Relaxation Effects in Nuclear Magnetic Resonance Absorption"; Physical Review, Vol. 73, Apr, 1948, pp. 679-712. Searchable Page
- NMR Frequency Table
Other reprints and reference materials can be found on the Physics 111 Library Site